The Colon Hydrotherapy System is a very simple, safe, comfortable, and hygienic way to irrigate the colon for endoscopic, radiological, or other medical purposes. The following sequence of events briefly describes the colon irrigation procedure. The tank is filled with hot and cold water controlled by a water mixing valve on the basin right topside. The proper temperature of the water falls between 99 -103 degrees Fahrenheit and is monitored by the temperature gauge visible on the front of the tower cabinet. If at any time the water rises above 103 degrees Fahrenheit, the temperature sensor will trigger the flow valve to shut off, stopping water flow to the client. After the client has comfortably cover themselves with a provided sheet and are positioned on the basin (with the pillow and cushioned backrest in place) and self-inserted the nozzle into their rectum, water flow can be safely activated by turning ON the FLOW switch. Water flow can be stopped at any time by simply turning the FLOW switch OFF. The tank line to the basin nipple has a backflow prevention valve as a permanent plumbing safety feature. It is located underneath the fiberglass basin and is connected directly to the basin nipple through an opening in the basin wall. It prevents water from flowing back into the line once it has passed through the basin nipple into the nozzle and into the client’s rectum. The System utilizes the controlled pressure created by gravity as water exits the elevated tank in the tower cabinet as activated by the FLOW switch. The procedure can be stopped at any time by simply turning off the FLOW switch to stop the gravity-fed stream of water or by the person simply sliding back off the nozzle and resting on the basin. A typical colon hydrotherapy session lasts approximately 30-40 minutes and uses approximately 10 gallons of water, which is the total volume capacity of the main tank. Usually, only one tank full of water is needed. At the end of the session, all three switches (U.V., CYCLE, FLOW) are turned OFF and the client slides back off the nozzle so that it naturally drops away from his body. The client can now drain comfortably on the ergonomic basin, sitting up erect or leaning back in a lounge position to fully drain her colon. After this is done, the client can rinse using the personal shower sprayer and its control valve located on the basin left topside. Tempered water is delivered to the sprayer by the same water mixing valve on the right topside of the basin that also sends water to the System tank. The client can then towel themselves dry before dressing and leaving the privacy of the colon irrigation room. follow url Is colon therapy right for me? Click here for more information in this PDF.